All posts 1 day 7 days 2 weeks 1 month 3 months 6 months 1 year Sort by: Could you please help me to correct the program if something is wrong? If you have that kind of adaptor, you can use a null-modem adaptor to switch them back or use a different usb adaptor Message 2 of I do not know if my program is not correct or there is problem with the cable.
|Date Added:||8 January 2010|
|File Size:||35.36 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
So if I can use software trigger it would be good. I want to pulse the voltage between 0V and some desired voltage for a set amount of time.
Otherwise why not just use the 2x and have them source and measure at the same time? I need keithley labview 50 readings per second.
Sample Rate with Labview – Tektronix and Keithley
I even tried deleting the graph from the loop and it was still reaching the same limitations. No registered users and keithley 6485 labview guests. Keithley 6485 labview anyone give me some advice? I do not know if my program is not correct or there is problem with the cable.
Labviea the manual, it requires to use a straight-through cable RS Message 2 of I am trying to use a to source voltage and measure current as a function of time.
I have set up the keithley labview command in a loop that is controlled by a wait timer. You can use only one interface at a time. Message 8 of If you have that kind of adaptor, you can use a null-modem adaptor to switch them back or use a different usb adaptor I made up the number 50 — keithley 6485 labview sure none of the responses are longer than whatever you set!
I do this with instruments when exact measurement timing isn’t an issue.
Tektronix and Keithley
Select “Keithley” from the “Manufacturer” keithley 6485 labview, and type “” into the “Additional keywords” field. I am only referring to the i-t measurement folder, and not the I-V scan. Taking a keithley 6485 labview keithleey at the manual for the KeithleyI read these sections: Author Post time Subject Direction: Tell us if you get any error msg, so we can guide you further.
I forgot to limit the keithley 6485 labview of the While loop in my example, this is usually wise to do, see the snippet below where the reading set to 1 sec speed Discover the different configuration VIs for your unit. Message 6 of Message 4 of 5.
If you need an earlier copy, just ask or recreate from the snippet below. As was pointed out the Keithley driver Lavbiew will be very helpful.
Tell us if you get any error msg, so we can guide you further. Message 5 of Could you please help keithley 6485 labview to correct the program if something is wrong? I will try the steps your suggested. At the factory, the GPIB is keithley 6485 labview. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
There was a error out. Perhaps you could try this VI?
Keithley Picoammeter – Instrument Driver – National Instruments
I am thinking the is taking too long to make the measurement. Or, post back if the one you think will do does.
You could simplify your keithley 6485 labview with a pair of Keithley s, s or s. Message 7 of Did you set the mode to RS? Is it possible that the limiting factor could be referencing the dynamic library where as your program just uses kdithley direct call to the instrument over GPIB?